Statement of problem
Anterior cantilever resin-bonded prostheses fail as a result of a labio-lingual peeling action, which creates a stress concentration within the adhesive layer.
The purpose of this study was to identify the factors that determine the retention of an anterior resin-bonded prosthesis and to seek to eliminate the stress concentration within the adhesive layer by fundamentally altering the prosthesis design.
Material and methods
The first experiment involved 40 Ni/Cr (Wiron 99) beams with a width of 5 mm, thickness of 0.5 mm, and lengths ranging from to 13 to 22 mm. The beams were cemented onto a block of the same material using an adhesive resin luting agent (Panavia 21). The length of the beam that was bonded ranged from 1 to 10 mm, resulting in a bonded area ranging from 5 to 50 mm2. A load was applied onto the cantilevered portion of the beam 2 mm from the end, causing a peeling action. The force (N) required to debond these beams was measured using a pull-to-fracture test. Subsequently, a second experiment was undertaken, and 7 beams with an altered point of attachment (new design) were tested. The new design had the point of attachment of the cantilevered portion located centrally on the bonded area of the beam. Implementing this new design clinically would result in a cantilevered resin-bonded fixed partial denture that would have the connector arm attached more centrally on the retainer wing. The data were analyzed using a 1-way analysis of variance (α=.05), and a Tukey pairwise comparison test was used when the results was statistically significant. Two finite element analysis (FEA) models, one simulating the first experimental design and the other simulating the new design, were created. A load was then applied on the cantilevered portion of the beams similar to the experimental models, and the stress patterns were examined. The numerical values of these resultant stresses were plotted graphically.
The direction of load application, which may be transferred to a clinical setting as labio-lingual forces, was identified as the dominant force responsible for debonding. The new design, which addressed this problem, showed a significant increase (P<.001) in retention. The FEA models identified the stress concentrations within the adhesive layer of the traditional design, which were eliminated when the new design was tested.
For the in vitro model, loads that may be interpreted clinically as labio-lingual forces resulted in the lowest forces required to cause debonding, and these forces were independent of the surface area of bonding. Altering the point of attachment of the cantilevered portion onto the retainer caused a significant increase in the forces needed to cause debonding.
To read this article in full you will need to make a payment
Purchase one-time access:Academic & Personal: 24 hour online accessCorporate R&D Professionals: 24 hour online access
One-time access price info
- For academic or personal research use, select 'Academic and Personal'
- For corporate R&D use, select 'Corporate R&D Professionals'
Subscribe:Subscribe to Journal of Prosthetic Dentistry
Already a print subscriber? Claim online access
Already an online subscriber? Sign in
Register: Create an account
Institutional Access: Sign in to ScienceDirect
- Attachment of a splint to enamel of lower anterior teeth.J Prosthet Dent. 1973; 30: 418-423
- The clinical performance of cantilevered resin-bonded bridgework.J Dent. 1996; 24: 251-256
- An analysis of clinical studies on resin-bonded bridges.J Dent Res. 1991; 70: 146-149
- Cast metal, resin-bonded prostheses: A 10-year retrospective study.J Prosthet Dent. 1989; 61: 436-441
- Performance of 400 adhesive bridges fitted in a restorative dentistry department.J Dent. 1991; 19: 221-225
- Long-term survival characteristics of 832 resin retained bridges and splints provided in a post-graduate teaching hospital between 1978 and 1993.J Oral Rehabil. 1999; 26: 302-320
- Design principles for cantilevered resin-bonded fixed partial dentures.Quintessence Int. 2000; 31: 613-619
- Effects of groove placement on retention/resistance of maxillary anterior resin-bonded retainers.J Prosthet Dent. 1995; 74: 133-139
- A longitudinal clinical study of bonded fixed partial dentures: the first 5 years.J Prosthet Dent. 1991; 66: 336-342
- A long-term prospective study of the etched-cast restoration.Int J Prosthodont. 1993; 6: 428-434
- Clinical factors affecting adhesive fixed partial dentures: a 6-year study.J Prosthet Dent. 1993; 70: 300-307
- Long-term survival data from a clinical trial on resin-bonded bridges.J Dent. 1997; 25: 239-242
- Analysis of debond rates of resin-bonded prostheses.J Dent Res. 1993; 72: 1244-1248
- Tensile peel failure of resin-bonded Ni/Cr beams: an experimental and finite element study.J Dent. 1994; 22: 252-256
- Finite element analysis studies of a metal-ceramic crown on a first premolar tooth.Int J Prosthodont. 2002; 15: 521-527
- Comparison of photo-activation versus chemical or dual-curing of resin-based luting cements regarding flexural strength, modulus and surface hardness.J Oral Rehabil. 2001; 28: 1022-1028
- Etched castings: an improved retentive mechanism for resin-bonded retainers.J Prosthet Dent. 1982; 47: 52-58
- Micromorphological changes in resin-dentin bonds after 1 year of water storage.J Biomed Mater Res. 2002; 63: 306-311
University of Sheffield, Sheffield, United Kingdom
Presented at the British Society for Dental Research (BSDR) Annual Conference, Birmingham, UK, April 2004.
© 2006 The Editorial Council of The Journal of Prosthetic Dentistry. Published by Elsevier Inc. All rights reserved.