Statement of problem.
In some instances, an implant needs to be placed deep subgingivally, which may result in a less accurate impression of the implant.
The purpose of this study was to evaluate the effect of subgingival depth of implant placement on the accuracy of implant impressions.
Material and methods.
A stone master model was fabricated with 5 implant analogs (RN synOcta analog), embedded parallel to each other, at the center (E) and the 4 corners (A, B, C, and D). The vertical position of the shoulders of the implants was intentionally different among the implants: A and E were flush with the top surface of the model; B was 2 mm below, and C and D were 4 mm below the surface. The horizontal distances of implants A, B, C, and D from E were measured with a measuring microscope. A cross-shaped metal measuring bar was then fabricated and connected to E, with the arms of the casting designed to be 2 mm above the top surface of the model and incorporating a reference mark. With the measuring bar connected to E, the vertical distances from the apical surface of A, B, C, and D to the measuring reference marks were measured with a digital micrometer. The body of the impression coping for implant D was modified by adding 4 mm of additional impression coping, while standard impression copings were used for all other implants. Open tray impressions were made using medium-body polyether material (Impregum Penta) or a combination of putty and light-body vinyl polysiloxane (VPS) material (Elite HD+) (n=15). Then casts were poured with type IV dental stone. The vertical and horizontal distances of the casts were measured with the methods outlined above for the master model. The distortion values that were determined as differences between the measurements of the master model and those of the casts were collected for statistical analysis. Two-way and 1-way repeated measures ANOVA followed by Tukey's HSD test were performed to compare the distortion values (α=.05).
For vertical measurements, 2-way repeated measures ANOVA showed no significant depth (P=.36), material (P=.24), or interaction effects (P=.06). However, it showed significant depth effect for horizontal measurements (P=.01). Within the polyether group, 1-way repeated measures ANOVA showed significant differences in horizontal measurements among the implants with different depths (P=.03). The post hoc Tukey's test showed that the impression of 4-mm-deep implants with normal impression copings (C) was significantly less accurate than impressions of 0-mm-deep implants (A) (P=.02). Within the VPS group, there was no significant difference among the implants with different depths (P=.09).
There was no effect of implant depth on the accuracy of the VPS group. However, for the polyether group, the impression of an implant placed 4 mm subgingivally showed a greater horizontal distortion compared to an implant placed more coronally. Adding a 4-mm extension to the retentive part of the impression coping eliminated this difference. (J Prosthet Dent 2008;99:107-113)
To read this article in full you will need to make a payment
Purchase one-time access:Academic & Personal: 24 hour online accessCorporate R&D Professionals: 24 hour online access
One-time access price info
- For academic or personal research use, select 'Academic and Personal'
- For corporate R&D use, select 'Corporate R&D Professionals'
Subscribe:Subscribe to Journal of Prosthetic Dentistry
Already a print subscriber? Claim online access
Already an online subscriber? Sign in
Register: Create an account
Institutional Access: Sign in to ScienceDirect
- The glossary of prosthodontic terms.J Prosthet Dent. 2005; 94: 44
- Three-dimensional analysis of the casting accuracy of one-piece, osseointegrated implant-retained prostheses.Int J Prosthodont. 1993; 6: 346-363
- Tightening characteristics for screwed joints in osseointegrated dental implants.J Prosthet Dent. 1994; 71: 592-599
- The accuracy of three implant impression techniques: a three-dimensional analysis.Int J Oral Maxillofac Implants. 1994; 9: 533-540
- Measuring fit at the implant prosthodontic interface.J Prosthet Dent. 1996; 75: 314-325
- One clinical visit for a multiple implant restoration master cast fabrication.J Prosthet Dent. 1997; 78: 550-553
- Strategies to achieve fit in implant prosthodontics: a review of the literature.Int J Prosthodont. 1999; 12: 167-178
- The significance of passive framework fit in implant prosthodontics: current status.Implant Dent. 2001; 10: 85-92
- Experimental breakdown of peri-implant and periodontal tissues. A study in the beagle dog.Clin Oral Implants Res. 1992; 3: 9-16
- Different types of inflammatory reactions in peri-implant soft tissues.J Clin Periodontol. 1995; 22: 255-261
- Microbial findings of deep peri-implant bone defects.Int J Oral Maxillofac Implants. 1997; 12: 106-112
- Microbiota associated with experimental peri-implantitis and periodontitis in adult Macaca mulatta monkeys.J Periodontol. 1998; 69: 190-194
- Microbial findings at failing implants.Clin Oral Implants Res. 1999; 10: 339-345
- Clinical methods for evaluating implant framework fit.J Prosthet Dent. 1999; 81: 7-13
- Loose gold screws frequently occur in full-arch fixed prostheses supported by osseointegrated implants after 5 years.Int J Oral Maxillofac Implants. 1994; 6: 169-178
- The response of bone in primates around unloaded dental implants supporting prostheses with different levels of fit.J Prosthet Dent. 1996; 76: 500-509
- Prosthesis misfit and marginal bone loss in edentulous implant patients.Int J Oral Maxillofac Implants. 1996; 11: 620-625
- Effect of prosthetic superstructure accuracy on the osteointegrated implant bone interface.Oral Surg Oral Med Oral Pathol Oral Radiol Endod. 1997; 83: 198-205
- Bone response to implant-supported frameworks with differing degrees of misfit preload: in vivo study in rabbits.Clin Implant Dent Relat Res. 2000; 2: 129-137
- Accuracy of open tray implant impressions: an in vitro comparison of stock versus custom trays.J Prosthet Dent. 2003; 89: 250-255
- The accuracy of implant verification casts compared with casts produced from a rigid transfer coping technique.J Prosthodont. 1996; 5: 248-252
- Accuracy of implant impression splinted techniques: effect of splinting material.Int J Oral Maxillofac Implants. 1999; 14: 885-888
- Evaluation of impression accuracy for osseointegrated implant supported superstructures.J Prosthet Dent. 2000; 83: 555-561
- In vitro comparison of master cast accuracy for single-tooth implant replacement.J Prosthet Dent. 2000; 83: 562-566
- A laboratory investigation of the accuracy of two impression techniques for single-tooth implants.Int J Prosthodont. 2001; 14: 152-158
- Evaluation of the accuracy of three techniques used for multiple implant abutment impressions.J Prosthet Dent. 2003; 89: 186-192
- Accuracy of 2 impression techniques for ITI implants.Int J Oral Maxillofac Implants. 2004; 19: 517-523
- Evaluation of transfer impressions for osseointegrated implants at various angulations.Implant Dent. 2004; 13: 358-366
- Evaluation of the accuracy of 3 transfer techniques for implant-supported prostheses with multiple abutments.Int J Oral Maxillofac Implants. 2004; 19: 192-198
- An evaluation of impression techniques for multiple internal connection implant prostheses.J Prosthet Dent. 2004; 92: 470-476
- Master cast accuracy in single-tooth implant replacement cases: an in vitro comparison. A technical note.Int J Oral Maxillofac Implants. 2005; 20: 455-460
- Impression techniques and misfit-induced strains on implant-supported superstructures: an in vitro study.Int J Periodontics Restorative Dent. 2006; 26: 379-385
- Verification jig for implant-supported prostheses: A comparison of standard impressions with verification jigs made of different materials.J Prosthet Dent. 2002; 88: 329-336
- Intraclass correlations: uses in assessing rater reliability.Psychol Bull. 1979; 86: 420-428
- Comparison of impression materials for direct multi-implant impressions.J Prosthet Dent. 2000; 83: 323-331
- Temperature effects on the rheological properties of current polyether and polysiloxane impression materials during setting.J Prosthet Dent. 2003; 90: 150-161
- Mechanical properties of 3 hydrophilic addition silicone and polyether elastomeric impression materials.J Prosthet Dent. 2004; 92: 151-154
- Clinically relevant mechanical properties of elastomeric impression materials.Int J Prosthodont. 1998; 11: 219-223
- Tolerance measurements of various implant components.Int J Oral Maxillofac Implants. 1997; 12: 371-375
- Comparison of interface relationships between implant components for laser-welded titanium frameworks and standard cast frameworks.Int J Oral Maxillofac Implants. 1999; 14: 491-495
- Displacement of implant components from impressions to definitive casts.Int J Oral Maxillofac Implants. 2006; 21: 747-755
© 2008 The Editorial Council of the Journal of Prosthetic Dentistry. Published by Elsevier Inc. All rights reserved.