Statement of problem
The literature provides limited information regarding the performance of Ti-6Al-4V laser and plasma joints welded in prefabricated bars in dental applications.
The purpose of this study was to evaluate the mechanical strength of different diameters of Ti-6Al-4V alloy welded with laser and plasma techniques.
Material and methods
Forty-five dumbbell-shaped rods were created from Ti-6Al-4V and divided into 9 groups (n=5): a control group with 3-mm and intact bars; groups PL2.5, PL3, PL4, and PL5 (specimens with 2.5-, 3-, 4-, and 5-mm diameters welded with plasma); and groups L2.5, L3, L4, and L5 (specimens with 2.5-, 3-, 4-, and 5-mm diameters welded with laser). The specimens were tested for ultimate tensile strength (UTS), and elongation percentages (EP) were obtained. Fractured specimens were analyzed by stereomicroscopy, and welded area percentages (WAP) were calculated. Images were made with scanning electron microscopy. In the initial analysis, the data were analyzed with a 2-way ANOVA (2×4) and the Tukey Honestly Significant Difference (HSD) test. In the second analysis, the UTS and EP data were analyzed with 1-way ANOVA, and the Dunnett test was used to compare the 4 experimental groups with the control group (α=.05). The Pearson and Spearman correlation coefficient tests were applied to correlate the study factors. Finite element models were developed in a workbench environment with boundary conditions simulating those of a tensile test.
The 2-way ANOVA showed that the factors welding type and diameter were significant for the UTS and WAP values. However, the interaction between them was not significant. The 1-way ANOVA showed statistically significant differences among the groups for UTS, WAP, and EP values. The Dunnett test showed that all the tested groups had lower UTS and EP values than the control group. The 2.5- and 3-mm diameter groups showed higher values for UTS and WAP than the other test groups. A positive correlation was found between welded area percentage and UTS and a negative correlation between these parameters and the diameters of the specimens. No statistically significant difference was found between the weld techniques.
Under the experimental conditions described, diameters of 2.5 and 3 mm resulted in higher UTS and WAP for both laser and plasma welding and appear to be the best option for joining prefabricated rods in this kind of union.
To read this article in full you will need to make a payment
Purchase one-time access:Academic & Personal: 24 hour online accessCorporate R&D Professionals: 24 hour online access
One-time access price info
- For academic or personal research use, select 'Academic and Personal'
- For corporate R&D use, select 'Corporate R&D Professionals'
Subscribe:Subscribe to Journal of Prosthetic Dentistry
Already a print subscriber? Claim online access
Already an online subscriber? Sign in
Register: Create an account
Institutional Access: Sign in to ScienceDirect
- New approach for immediate prosthetic rehabilitation of the edentulous mandible with three implants: a retrospective study.Clin Oral Implants Res. 2011; 22: 1265-1269
- Phillips’ science of dental materials.12th ed. Elsevier, Saunders, St. Louis2013: 367-395
- Comparative analysis of different joining techniques to improve the passive fit of cobalt-chromium superstructures.J Prosthet Dent. 2012; 108: 377-385
- Fatigue strength in laser welding of the lap joint.Finite Elem Anal Des. 2004; 40: 1059-1070
- Laser welding of Ti6Al4V titanium alloy.J Mater Proc Technol. 2009; 209: 3705-3713
- Laser welding of titanium in dentistry.Acta Odontol Scand. 1988; 46: 247-253
- Mechanical properties of laser-welded cast and wrought titanium.J Prosthet Dent. 1995; 74: 250-257
- Joining titanium materials with tungsten inert gas welding, laser welding, and infrared brazing.J Prosthet Dent. 1995; 74: 521-530
- Mechanical properties of titanium connectors.Int J Prosthodont. 1996; 9: 379-393
- Mechanical properties of laser-welded cast titanium joints under different conditions.J Prosthet Dent. 1998; 79: 477-483
- Effects of joint configuration for the arc welding of cast Ti-6Al-4V alloy rods in argon.J Prosthet Dent. 1998; 79: 291-297
- Flexural strength of pure Ti, Ni-Cr and Co-Cr alloys submitted to Nd:YAG laser or TIG welding.Braz Dent J. 2006; 17: 20-23
- Laser welding of cast titanium and dental alloys using argon shielding.J Prosthodont. 2006; 15: 102-107
- Effect of laser irradiation conditions on the laser welding strength of cobalt-chromium and gold alloys.J Oral Sci. 2011; 53: 301-305
- Effects of laser-weld joint opening size on fatigue strength of Ti-6Al-4V structures with several diameters.J Oral Rehabil. 2011; 38: 196-201
- Intraoral laser welding: ultrastructural and mechanical analysis to compare laboratory laser and dental laser.Lasers Med Sci. 2011; 26: 415-420
- Studies on laser- and plasma-welded titanium.Dent Mater. 1993; 9: 104-107
- Joint strength of laser-welded titanium.Dent Mater. 2002; 18: 143-148
- Corrosion-fatigue of laser-repaired commercially pure titanium and Ti-6Al-4V alloy under different test environments.J Oral Rehabil. 2004; 31: 1029-1034
- Penetration depth into dental casting alloys by Nd:YAG laser.J Biomed Mater Res B Appl Biomater. 2005; 72: 64-68
- Evaluation of welded titanium joint used with cantilevered implant-supported prostheses.J Prosthet Dent. 2006; 96: 25-36
- Fracture resistance of Na:YAG Laser-welded cast titanium joints with various clinical thicknesses and welding pulse energies.Dent Mater J. 2007; 26: 367-372
- Comparative study on torsional strength, ductility and fracture characteristics of laser-welded alpha+beta Ti-6Al-7Nb alloy, CP titanium and Co-Cr alloy dental castings.Dent Mater. 2008; 24: 839-845
- Welding techniques in dentistry.in: Kovacevic R. Welding Processes. 17th ed. In tech, Croatia2012: p.415-p.438
- Effects of soldering methods on tensile strength of a gold-palladium metal ceramic alloy.J Prosthet Dent. 2014; 112: 994-1000
- Standard methods for mechanical testing of welds. ANSI/AWS B4.0-92. Specification: E 8M – 04: Standard test methods for tension testing of metallic materials.American Welding Society, Miami1998: 15-23
- Understanding stainless steel heat-affected zones.Welding J. 2005; : 26-30
- Properties and sulfide stress cracking resistance of coarse-grained heat-affected zones in V-microalloyed X60 steel pipe.Welding J. 2005; : 113-123
Published online: March 04, 2015
This study is supported by the National Council for Scientific and Technological Development (CNPQ) (07/2010 PIBIC/CNPq/UFU) and the Foundation for Research of the State of Minas Gerais (FAPEMIG) (SAU19/20009).
© 2015 Editorial Council for the Journal of Prosthetic Dentistry. Published by Elsevier Inc. All rights reserved.