Statement of problem
The selection of an appropriate restorative material based on fracture behavior is important for the marginal integrity of a dental restoration. For computer-aided design and computer-aided manufacturing (CAD-CAM) restorative materials, information regarding their edge chipping resistance is scarce.
The purpose of this in vitro study was to determine the edge chipping resistance (ECR) and Martens hardness (HM) of 6 different dental CAD-CAM restorative materials before and after thermocycling.
Material and methods
Four composite resin materials including Brilliant Crios; Cerasmart, an experimental material; Lava Ultimate, a polymer-infiltrated ceramic-network (PICN) material (VITA Enamic), and a glass-ceramic control (IPS Empress CAD) were tested. The specimens were tested before and after thermocycling (30 000 times, 5 °C/55 °C). The ECR was measured for each material (n=25) and related to the point of loading and to the maximum chipping depth. HM was determined for each material (n=25). The Kruskal-Wallis and Mann-Whitney U tests were used to compare materials (α=.05). The impact of thermocycling was analyzed by using the Wilcoxon test (α=.05). The correlations between all parameters were calculated by using the Spearman-Rho test (α=.05). For fractographic analysis of chip patterns, chipped surfaces were analyzed by laser scanning microscopy.
For ECR and HM, the materials showed different values. ECRmd and ECRpl showed a positive correlation, but both showed a negative correlation to HM. The materials showed a different chip size (P<.001). Chip patterns revealed brittle material behavior in all cases.
All tested CAD-CAM materials behaved as brittle materials, but HM and ECR differed among the materials. The control glass-ceramic material showed the highest values for HM, followed by the PICN material. ECR values revealed the opposite order of materials, with the highest for composite resins. Artificial aging by thermocycling affected all dental CAD-CAM restorative materials. Especially for composite resin materials, ECR changed after aging.
To read this article in full you will need to make a payment
Purchase one-time access:Academic & Personal: 24 hour online accessCorporate R&D Professionals: 24 hour online access
One-time access price info
- For academic or personal research use, select 'Academic and Personal'
- For corporate R&D use, select 'Corporate R&D Professionals'
Subscribe:Subscribe to Journal of Prosthetic Dentistry
Already a print subscriber? Claim online access
Already an online subscriber? Sign in
Register: Create an account
Institutional Access: Sign in to ScienceDirect
- Factors essential for successful all-ceramic restorations.J Am Dent Assoc. 2008; 139: 14S-18S
- Evaluation of five CAD/CAM materials by microstructural characterization and mechanical tests: a comparative in vitro study.BMC Oral Health. 2018; 18: 5
- Dental biomaterials for chairside CAD/CAM: state of the art.J Adv Prosthodont. 2017; 9: 486-495
- Physicomechanical characterization of polyetheretherketone and current esthetic dental CAD/CAM polymers after aging in different storage media.J Prosthet Dent. 2016; 115: 321-328
- Chairside CAD-CAM materials. Part 1: measurement of elastic constants and microstructural characterization.Dent Mater. 2017; 33: 84-98
- Chairside CAD/CAM materials. Part 2: flexural strength testing.Dent Mater. 2017; 33: 99-109
- Evaluation of mechanical and optical behavior of current esthetic dental restorative CAD/CAM composites.J Mech Behav Biomed Mater. 2015; 55: 1-11
- Load-bearing capacity of CAD/CAM milled polymeric three-unit fixed dental prostheses: effect of aging regimens.Clin Oral Investig. 2012; 16: 1669-1677
- Mechanical properties and internal fit of 4 CAD-CAM block materials.J Prosthet Dent. 2018; 119: 384-389
- From artinasal to CAD-CAM blocks: state of the art of indirect composites.J Dent Res. 2016; 95: 487-495
- Degradation of polymeric restorative materials subjected to a high caries challenge.Dent Mater. 2011; 27: 244-252
- Are linear elastic material properties relevant predictors of the cyclic fatigue resistance of dental resin composites?.Dent Mater. 2014; 30: 381-391
- Fracture toughness analysis of ceramic and resin composite CAD/CAM materials.Oper Dent. 2019; 44-4: E190-E201
- Filler characteristics of modern dental resin composites and their influence on physico-mechanical properties.Dent Mater. 2016; 32: 1586-1599
- Factors involved in mechanical fatigue degradation of dental resin composites.J Dent Res. 2013; 92: 584-591
- Springer handbook of metrology and testing.Springer-Verlag, Berlin Heidelberg2011: 381
- The effect of artificial aging on Martens hardness and indentation modulus of different dental CAD-CAM restorative materials.J Mech Behav Biomed Mater. 2018; 86: 191-198
- Edge-strength of resin-composite margins.Dent Mater. 2008; 24: 129-133
- Implant-supported full-arch zirconia-based mandibular fixed dental prostheses. Eight-year results from a clinical pilot study.Acta Odontol Scand. 2012; 71: 1118-1122
- Clinical efficacy of veneered zirconium dioxide-based posterior partial fixed dental prostheses: five-year results.J Prosthet Dent. 2012; 108: 214-222
- Prospective evaluation of zirconia posterior fixed partial dentures: four-year clinical results.Int J Prosthodont. 2010; 23: 141-148
- Edge-strength of flowable resin-composites.J Dent. 2008; 36: 63-68
- Fractography of ceramics and glasses.National Institute of Standards and Technology, Washington2016 (10-31, 4-40, 4-42)
- A universal relation for edge chipping from sharp contacts in brittle materials: a simple means of toughness evaluation.Acta Mater. 2007; 55: 2555-2561
- Edge strength of CAD/CAM materials.J Dent. 2018; 74: 95-100
- Chipping fracture resistance of denture tooth materials.Dent Mater. 2014; 30: 545-553
- Chipping fracture resistance of dental CAD/CAM restorative materials: part I, procedures and results.Dent Mater. 2014; 30: e99-e111
- Criteria of ceramics fracture (edge chipping and fracture toughness tests).Ceram Int. 2013; 39: 3293-3300
- Unified curve of the edge chipping resistance in connection with the rounding radius indenter.Eng Fract Mech. 2017; 178: 265-278
- Edge chipping test on microhybrid and nanocomposites.Dent Mater. 2013; 29S: e61
- ISO CEN/TS 843-9:2010. Advanced technical ceramics - mechanical properties of monolithic ceramics at room temperature - Part 9: method of test for edge-chip resistance.International Organization for Standardization, Geneva2010 (Available at:)
- Ceramic toughness assessment through edge chipping measurements – influence of interfacial friction.J Eur Ceram Soc. 2009; 29: 2135-2141
- On the chipping and splitting of teeth.J Mech Behav Biomed Mater. 2011; 4: 315-321
- On edge chipping in cylindrical surfaces.Int J Solid Struct. 2015; 54: 12-19
- Classification of ceramics and glass (edge chipping and fracture toughness).Ceram Int. 2014; 40: 5591-5596
- Edge chipping resistance of ceramics: problems of test method.J Adv Ceram. 2013; 2: 370-377
- In vitro study of edge-strength of provisional polymer-based crown and fixed partial denture materials.Dent Mater. 2007; 23: 1570-1573
- Edge strength of indirect restorative materials.J Dent. 2009; 37: 799-806
- ISO/TS11405:2015. Dentistry – testing of adhesion to tooth structure.International Organization for Standardization, Geneva2015 (Available at:)
- Thermal cycling procedures for laboratory testing of dental restorations.J Dent. 1999; 27: 89-99
- ISO 14577-1:2002. Metallic materials – instrumented indentation test for hardness and material parameters. Part 1. Test method.International Organization for Standardization, Geneva2002 (Available at:)
Published online: February 27, 2020
© 2020 by the Editorial Council for The Journal of Prosthetic Dentistry.