Statement of problem
Although several manufacturers market soft metal milling blanks and systems, comprehensive comparative studies of differences in properties across commercially available soft metal milling alloys are lacking.
The purpose of this in vitro study was to compare the microstructures and mechanical properties of 3 soft metal milling cobalt-chromium (Co-Cr) alloys (Ceramill Sintron, Soft Metal, and Sintermetall).
Material and methods
Disk-shaped specimens (for surface characterization and hardness test) and dumbbell-shaped specimens (for tensile test as per International Organization for Standardization (ISO) 22674) were prepared by following each soft metal milling manufacturer’s instructions. The crystal structures and microstructures of the 3 alloys were evaluated with optical microscopy, X-ray diffractometry (XRD), and scanning electron microscopy with electron backscattered diffraction (EBSD). The mechanical properties were investigated with a tensile test and Vickers hardness test (n=6). The results of the mechanical (tensile and hardness) tests were analyzed with 1-way ANOVA and the post hoc Tukey multiple comparison test (α=.05).
The Sintermetall specimen showed a finer microstructure and more porosity than the other 2 alloys. The XRD and EBSD analyses showed that the γ (face-centered cubic, fcc) matrix phase was predominant in the Ceramill Sintron alloy and the ε (hexagonal close-packed, hcp) matrix phase was predominant in the Soft Metal alloy. The Sintermetall alloy showed a slightly higher amount of ε phase than γ phase, with more chromium carbide formation than the other 2 alloys. The Ceramill Sintron alloy showed a significantly higher tensile strength than the other 2 alloys (P<.05), but a significantly lower 2% offset yield strength than the other 2 alloys (P<.05). The highest elongation was found in the Ceramill Sintron alloy, followed by the Sintermetall and Soft Metal alloys. The elastic modulus was the highest in the Sintermetall alloy, followed by the Soft Metal and Ceramill Sintron alloys. No significant differences in Vickers hardness values were detected among the 3 alloys (P=.263).
The different commercially available soft metal milling blanks and systems produced dissimilar alloys in terms of crystal structures and microstructures and, as a result, different mechanical properties.
To read this article in full you will need to make a payment
Purchase one-time access:Academic & Personal: 24 hour online accessCorporate R&D Professionals: 24 hour online access
One-time access price info
- For academic or personal research use, select 'Academic and Personal'
- For corporate R&D use, select 'Corporate R&D Professionals'
Subscribe:Subscribe to Journal of Prosthetic Dentistry
Already a print subscriber? Claim online access
Already an online subscriber? Sign in
Register: Create an account
Institutional Access: Sign in to ScienceDirect
- Marginal and internal adaptation of milled cobalt-chromium copings.J Prosthet Dent. 2015; 114: 680-685
- Internal and marginal fit of cobalt-chromium fixed dental prostheses fabricated with 3 different techniques.J Prosthet Dent. 2015; 114: 686-692
- A microcomputed tomography evaluation of the marginal fit of cobalt-chromium alloy copings fabricated by new manufacturing techniques and alloy systems.J Prosthet Dent. 2017; 117: 393-399
- In vitro evaluation of marginal, axial, and occlusal discrepancies in metal ceramic restorations produced with new technologies.J Prosthet Dent. 2016; 116: 368-374
- Marginal fit of metal-ceramic crowns fabricated by using a casting and two selective laser melting processes before and after ceramic firing.J Prosthet Dent. 2019; 122: 475-481
- Casting alloys.Dent Clin North Am. 2004; 48: 499-512
- Metal-ceramic alloys in dentistry: a review.J Prosthodont. 2009; 18: 188-194
- Enhanced biocompatibility of a Ni–Cr alloy prepared by selective laser melting: a preliminary in vitro study.J Mater Sci Technol. 2019; 8: 1587-1592
- Effect of carbon addition on microstructure and mechanical properties of a wrought Co–Cr–Mo implant alloy.Mater Trans. 2006; 47: 287-290
- Martensitic transformation in a cast Co-Cr-Mo-C alloy.Metall Mater Trans A. 2008; 39: 8-18
- Significant improvement in mechanical properties of biomedical Co-Cr-Mo alloys with combination of N addition and Cr-enrichment.Mater Trans. 2008; 49: 260-264
- Microstructures and mechanical properties of Co-29Cr-6Mo alloy fabricated by selective laser melting process for dental applications.J Mech Behav Biomed Mater. 2013; 21: 67-76
- Preliminary evaluation of mechanical properties of Co-Cr alloys fabricated by three new manufacturing processes.Int J Prosthodont. 2015; 28: 396-398
- Physico-mechanical properties and prosthodontic applications of Co-Cr dental alloys: a review of the literature.J Adv Prosthodont. 2014; 6: 138-145
- Fracture behavior of metal-ceramic fixed dental prostheses with frameworks from cast or a newly developed sintered cobalt-chromium alloy.Clin Oral Investig. 2015; 19: 401-411
- Sintering behaviour of Co-28%Cr-6%Mo compacted blocks for dental prosthesis.J Mater Res Tech. 2019; 8: 2052-2062
- Effect of post-sintering conditions on the mechanical properties of a new Co–Cr alloy produced by new subtractive manufacturing.J Nanosci Nanotechnol. 2019; 19: 2395-2398
- A novel CAD/CAM base metal compared to conventional CoCrMo alloys: an in-vitro study of the long-term metal-ceramic bond strength.Oral Health Dent Manag. 2014; 13: 446-452
- Effect of different post-sintering temperatures on the microstructures and mechanical properties of a pre-sintered Co–Cr alloy.Metals. 2018; 8: 1036
- Manufacturing technology for aerospace structural materials.1st ed. Elsevier, Oxford2006: 541-566
- ISO 22674:2016 dentistry - metallic materials for fixed and removable restorations and appliances.International Organization for Standardization, Geneva2016 (Available at:)
- Effects of chromium and nitrogen content on the microstructures and mechanical properties of as-cast Co-Cr-Mo alloys for dental applications.Acta Biomater. 2012; 8: 2856-2862
- Microstructures and mechanical properties of Co-Cr dental alloys fabricated by three CAD/CAM-based processing techniques.Materials. 2016; 9: 596
- Effects of grain size, texture and twinning on mechanical properties and work-hardening behavior of AZ31 magnesium alloys.Mater Sci Eng A. 2011; 528: 8537-8545
- Effect of grain sizes on mechanical properties and biodegradation behavior of pure iron for cardiovascular stent application.Biomatter. 2016; 6: e959874
- A quantitative evaluation of microstructure by electron back-scattered diffraction pattern quality variations.Microsc Microanal. 2013; 19: 83-88
- Microstructure, phase content, and thermal stability of a cast Co–Cr dental alloy after porcelain sintering cycles using electron backscatter diffraction.J Mater Res. 2015; 30: 2188-2196
- Comparison of the microstructure and phase stability of as-cast, CAD/CAM and powder metallurgy manufactured Co-Cr dental alloys.Dent Mater. 2015; 31: e306-e315
- A generalized self-consistent polycrystal model for the yield strength of nanocrystalline materials.J Mech Phys Solids. 2004; 52: 1125-1149
- Role of five-fold twin boundary on the enhanced mechanical properties of fcc Fe nanowires.Nano Lett. 2011; 11: 5264-5273
- Deformation induced FCC to HCP transformation in a Co-27Cr-5Mo-0.05C alloy.Mater Sci Eng A. 2011; 528: 3037-3043
- Cobalt-based orthopaedic alloys: relationship between forming route, microstructure and tribological performance.Mater Sci Eng C. 2012; 32: 1222-1229
Published online: December 07, 2020
Supported by the National Research Foundation of Korea (NRF) grant funded by the Korea government (MSIT) (2017R1A5A2015391).
© 2020 by the Editorial Council for the Journal of Prosthetic Dentistry.